Federal Interagency Sedimentation Project
US D-96 Depth integrating suspended hand line sampler
HIF/FISP Part # 4101018
OFA Price: $6,990.00*/Each
*Pricing may change due to manufacturer reordering,
please call the HIF at (800) 382-0634 ext. 83271 for final pricing and
The US D-96 is a collapsible-bag suspended-sediment/water-quality sampler capable of collecting a 3-liter sample. It is 35 in long, weighs 132 lbs, and has a hollow cavity inside the sampler body. It is fabricated from bronze and aluminum castings with a high-density polyethylene tail. All metal parts are plastic coated. The sampler employs a sliding tray that supports the bag and holds the nozzle holder with nozzle in place. The bag is attached to the nozzle holder with a hook-and-loop strap. The sampler uses a plastic or TFE nozzle and a plastic or PFA bag as the sample container. The US D-96 meets the protocol for water-quality sampling as outlined in the USGS’s National Field Manual for the Collection of Water-Quality Data. The sampler is protected by US Patent No. 6,216,549 B1.
The US D-96 sampler will collect acceptable flow-weighted samples in streams velocities ranging from 2.0 to 12.5 ft/sec. Extreme care should be practiced when deploying the sampler at stream velocities above 10 ft/sec. The US D-96 sampler is capable of sampling to a depth of 39 ft with a 5/16-in internal diameter nozzle, 60 ft with a 1/4 in internal diameter nozzle, and 110 ft with a 3/16-in internal diameter nozzle. The unsampled zone using the US D-96 sampler 4 in. HIF Repairable.
Where and how the sampler is deployed in the field is as important as the fundamental design of the sampler. Additional information about how to properly use this sampler can be found in the following references:
Wilde, F.D., Radtke, D.B., Gibs, J., and Iwatsubo, R.T., 1998, National Field Manual for the Collection of Water-Quality Data: U.S. Geological Survey Techniques of Water-Resources Investigations, Selection of Equipment for Water Sampling, book 9, chap. A27, available at: http://pubs.usgs.gov/twri/.
Operation instructions are given in Instruction for D-96 and also in the following reports:
4107003 ADAPTER SLEEVE, PINT